New fitting for use in (ultra) high pressure liquid chromatography systems made from VICTREX™ PAEK polymer
| Subj: Press-releses
VICI AG International, Schenkon/Switzerland, a leading manufacturer of valves and fittings for analytical, biomedical and biocompatible precision instruments, is the first company to offer a new one-piece fitting using VICTREX™ PAEK high-performance polymer. According to VICI, the cost-effective injection moulded VICTREX HT™ fitting can withstand the high pressures inherent in liquid chromatography systems and is, in fact, capable of withstanding pressures of up to 1,000 bar (14,500 psi), enabling more flexibility and easier handling.1
The pressures required in liquid chromatography have been rising steadily for over five years, given that the particles of the separation materials (usually modified silica gels) now frequently have a grain size of two micrometres or less. Consequently, substances within mixtures can be separated more quickly, with identical efficiency.
High-quality material — high-quality fitting
The greater strength of VICTREX HT polymer compared to unreinforced PEEK prevents the fitting from breaking apart when it is being tightened, since the carbon fibres are aligned in parallel by the injection process; this extra strength extends the service life of the fitting and allows several cycles of tightening and loosening of the fitting.1 In the past, this has only been possible in the high pressure range using conventional two-piece ‘nut and ferrule’ fittings or complicated one-piece fittings. The design of such one-piece fittings often means they are only suitable for a specific connection and usually just for single use.
“Assisted by Victrex and its material and processing expertise, we have developed and commercialised a new product in this segment to offer users greater flexibility, reusability and above all easier handling compared to steel screw fittings. Also, being manufactured by injection moulding, it is favourable in cost terms, too”, explains Dr Andre Hütz, product manager at VICI AG International. “We have performed tests showing that, when tightened by hand, the new VICTREX HT fittings can be used at 500 bar and above. When tightened with a tool, they can also be reused at least three times after first use at a test pressure of 950 bar.”1
1) VICI has confirmed that the performance claims are based upon testing it has performed. According to VICI test documentation, results depend on the level of tightening.
Further information on VICI at www.vici-jour.com; https://www.vici.com/; http://www.vici-dbs.com
Source: VICI AG International